Why Should You Attend:
While calibration, which is often delegated to an accredited external laboratory, determines whether a gage or instrument is accurate, MSA is often performed in-house on actual parts to quantify the gage's precision. This presentation will show how to perform MSA, and check and report the results.
Areas Covered in the Webinar:
Who Will Benefit:
This webinar will provide valuable assistance to personnel in all manufacturing companies. Employees who will benefit include:
William A. Levinson, P.E., is the principal of Levinson Productivity Systems, P.C. He is an ASQ Fellow, Certified Quality Engineer, Quality Auditor, Quality Manager, Reliability Engineer, and Six Sigma Black Belt. He is also the author of several books on quality, productivity, and management, of which the most recent is The Expanded and Annotated My Life and Work: Henry Ford's Universal Code for World-Class Success.
A gage or instrument must be both accurate and precise to do its intended job. A gage's accuracy, which is ensured by calibration, reflects the agreement of its average measurement with the actual dimension of a standard. Calibration does not, however, reflect a gage's precision; its ability to get the same measurement consistently. Measurement systems analysis (MSA), or gage reproducibility and repeatability (R&R) measures the precision.
Measurement systems analysis (MSA) returns a gage's repeatability and reproducibility in terms of standard deviations. Lower is better because a smaller standard deviation means less variation. Repeatability measures the gage's ability to get the same measurement from the same part when it is used by the same inspector. This variation is inherent in the gage and the measurement procedure. Reproducibility measures the dependence of the result on the inspector. If reproducibility variation is dependent, it means that inspector judgment (such as the definition of "finger tight" or how the inspector reads a dial gage) affects the outcome.
Combination of the repeatability and reproducibility return the gage's overall standard deviation, which is in turn reflected by its precision/tolerance (P/T) ratio. This should ideally be less than 10%. If the P/T ratio is too large, then it is necessary to improve it (e.g. with better methods or gages) and/or protect the customer from borderline nonconforming parts by guard banding.
Our refund policy is governed by individual products and services refund policy mentioned against each of offerings. However in absence of specific refund policy of an offering below refund policy will be effective.
Registrants may cancel up to two working days prior to the course start date and will receive a letter of credit to be used towards a future course up to one year from date of issuance. ComplianceOnline would process/provide refund if the Live Webinar has been cancelled. The attendee could choose between the recorded version of the webinar or refund for any cancelled webinar. Refunds will not be given to participants who do not show up for the webinar. On-Demand Recordings can be requested in exchange. Webinar may be cancelled due to lack of enrolment or unavoidable factors. Registrants will be notified 24hours in advance if a cancellation occurs. Substitutions can happen any time. On-Demand Recording purchases will not be refunded as it is available for immediate streaming. However if you are not able to view the webinar or you have any concern about the content of the webinar please contact us at below email or by call mentioning your feedback for resolution of the matter. We respect feedback/opinions of our customers which enables us to improve our products and services. To contact us please email [email protected] call +1-888-717-2436 (Toll Free).
PAYMENT METHOD: 100% Secure Transaction